Capacity: 15 l
EM container (yellow) with capacity 15 l.
For disinfection of processed medical materials, with tap for discharging disinfectant solution.
EM container consists of a cylindrical 15 l tank of yellow color, which bottom is elevated for 9 cm. The wall at the bottom on the one side has a cut above which a tap screwed in the bottom is placed. Inside, above the bottom, a grid is located at height 14 mm. Inner and external covers are available.
Before using, a container should be assembled. To do this, insert a tap in a tap cut, so that a resin sealing ring was outside, and place a nut on a tap thread from the inner side. Holding a nut with a thumb, screw the tap and install it with a discharge nozzle facing down. Put a grid on the bottom of a container.
- Materials provided for disinfection should be put in a yellow polyethylene packet, which has ready cuts on the bottom.
- Put a packet on a grid.
- Pour a disinfectant solution in a container, cover a packet with an inner cover from above. Put an external cover and wait for disinfection time according to an instruction.
- Upon disinfection, discharge a disinfectant solution through a tap.
- Open an external cover and by pressing on an inner cover try to squeeze disinfectant solution from a packet. Take out an inner cover and a packet with materials subjected to disinfection, for further utilization.
- Inner cover
- External cover